Frequent exposure to suboptimal temperatures in vaccine cold-chain system in India: results of temperature monitoring in 10 states
Manoj V Murhekar a, Srihari Dutta b, Ambujam Nair Kapoor c, Sailaja Bitragunta d, Raja Dodum e, Pramit Ghosh f, Karumanagounder Kolanda Swamy g, Kalyanranjan Mukhopadhyay h, Somorjit Ningombam i, Kamlesh Parmar j, Devegowda Ravishankar k, Balraj Singh l, Varsha Singh m, Rajesh Sisodiya n, Ramaratnam Subramanian g & Tana Takum e
a. National Institute of Epidemiology, Indian Council of Medical Research, R127, Tamil Nadu Housing Board, Ayapakkam, Ambattur, Chennai 600 077, India.
b. United Nations Children’s Fund, New Delhi, India.
c. Indian Council of Medical Research, New Delhi, India.
d. Directorate of Health, Medical and Familty Welfare, Government of Andhra Pradesh, Hyderabad, India.
e. Department of Health and Family Welfare, Government of Arunachal Pradesh, Papum Pare, India.
f. Department of Community Medicine, Medical College, Kolkata, India.
g. Directorate of Public Health, Government of Tamil Nadu, Chennai, India.
h. Directorate of Health and Family Welfare, Government of West Bengal, Kolkata, India.
i. Directorate of Health Services, Government of Manipur, Imphal, India.
j. Department of Health and Family Welfare, Government of Gujarat, Gandhinagar, India.
k. Department of Health and Family Welfare, Government of Karnataka, Bengaluru, India.
l. Department of Community Medicine and Epidemiology, Indira Gandhi Medical College, Shimla, India.
m. State Institute of Health and Family Welfare, Government of Bihar, Patna, India.
n. Directorate of Health Services, Government of Madhya Pradesh, Bhopal, India.
Correspondence to Manoj V Murhekar (e-mail: firstname.lastname@example.org).
(Submitted: 22 February 2013 – Revised version received: 01 June 2013 – Accepted: 12 June 2013 – Published online: 09 September 2013.)
Bulletin of the World Health Organization 2013;91:906-913. doi: http://dx.doi.org/10.2471/BLT.13.119974
Childhood immunization is one of the most cost-effective health interventions.1 Since most vaccines are sensitive to heat, an adequate cold-chain system often has to be created and maintained to preserve the quality of a vaccine before it is administered. Although emphasis has long been placed on avoiding high temperatures during vaccine storage and shipment,2 exposure to “subzero” temperatures – i.e. temperatures < 0 °C – can also damage and reduce the potency of the diphtheria, tetanus and pertussis (DPT), diphtheria and tetanus, tetanus toxoid, hepatitis B and pentavalent vaccines.3–9 The World Health Organization (WHO) recommends that all childhood vaccines except the oral polio vaccine be kept at 2–8 °C during their in-country distribution.10
In India, the Universal Immunization Programme targets 27 million infants and 30 million pregnant women every year.10 The focus of the programme is on the vaccination of children aged less than 1 year against six vaccine-preventable diseases – tuberculosis, diphtheria, tetanus, pertussis, polio and measles – and the vaccination of pregnant women against tetanus. In some states, vaccines against hepatitis B and Japanese encephalitis are also included in the programme. There is a five-level supply chain for vaccines in India. This chain begins with government medical supply depots (n = 4). These depots supply state vaccine stores (n = 35), which, in turn, supply regional vaccine stores (n = 116). Vaccine is sent from the regional stores to district vaccine stores (n = 626), which, in turn, supply the last links in the chain: the primary or community health centres that act as peripheral vaccine stores (n = 26 439).10 Note that, in terms of the cold chain, a region is a subdivision of a state, not vice versa. Within this chain, all vaccines except the oral polio vaccine should be stored at 2–8 °C – either in walk-in coolers or ice-lined refrigerators – and be transported in cold boxes with ice packs that have been allowed to reach 0 °C. Although most of the cold-chain equipment in India is maintained by dedicated “cold-chain technicians” who are employed by the state-level departments of health, a few states buy such maintenance from the private sector. Immunization services in the public sector are mostly provided at district hospitals, urban health centres and primary or community health centres. However, outreach vaccination sessions at health subcentres or in remote villages are also conducted, using vaccine transported from the nearest primary or community health centre.
In India, several studies have assessed the exposure of vaccines to high temperatures.11–13 However, there appears to be no information on vaccine exposure to subzero temperatures in the national cold-chain system. We therefore conducted a study to estimate the frequency of vaccine exposure to suboptimal temperatures – i.e. temperatures of < 0 or > 8 °C – at various levels in the cold chain.
Selection of study sites
The study was conducted, from February to August 2012, by following a WHO protocol for monitoring temperatures in a cold-chain system for vaccines.2 Ten Indian states – Bihar and West Bengal in the east of the country, Arunachal Pradesh and Manipur in the north-east, Himachal Pradesh in the north, Andhra Pradesh, Karnataka and Tamil Nadu in the south, Gujarat in the west, and the central state of Madhya Pradesh – were selected to represent the main geographical regions of India (Appendix A, available at: http://www.nie.gov.in/Bulletin2013/Murhekar/AppendixA.pdf). In each of these 10 states, the district that was most distant from the state capital and the one that was largest in terms of either population (n = 9) or number of peripheral vaccine stores (n = 1) were selected for study. In each of the 20 selected districts, two peripheral health facilities where vaccine is stored – generally the two that were most distant from the district vaccine store – were selected, to give four peripheral vaccine stores – one urban and three rural – for investigation in each study state. We consulted with the managers of the Universal Immunization Programme to ensure that the peripheral vaccine stores that we selected for study were both accessible and staffed.2 We also investigated the most distant outreach session run by each of the peripheral vaccine stores that we investigated. In each study state, one or two researchers were responsible for planning, coordinating and monitoring the study.
Forty test boxes – each of which contained a TRIX-8 electronic temperature recorder (LogTag Recorders, Auckland, New Zealand), two vials of DPT vaccine (procured from the state vaccine store for Tamil Nadu in Chennai), a FreezeAlert freeze-event indicator (Sensitech, Beverly, United States of America) and a TransTracker C freeze-event indicator (Temptime, Morris Plains, USA) – were prepared. The recorder used is a datalogger that records temperatures every 30 minutes. The DPT vials passed a shake test14 – indicating they had never been frozen – before they were placed in the boxes.
Four of these test boxes – one for each of the peripheral vaccine stores that we investigated in the state – were placed in the state vaccine store – or regional vaccine store if no state store existed – in each study state. Each of these boxes was then stored, handled and transported, exactly as if it contained routine supplies of DPT vaccine, to one of the peripheral vaccine stores that we investigated. Each box was stored for at least 1 month in a state vaccine store (if present), 1 month in a regional vaccine store (if present), 1 month in a district vaccine store and 2 weeks in a peripheral vaccine store. Although the card boxes used for the study were the same as those used for routine vaccine shipments in India, they could be identified by the personnel involved in the cold chain. Each test box was accompanied by a monitoring sheet that was used by the staff in each store – or the research teams – to record when the box entered the store and when it left.
Temperature monitoring during outreach sessions
For outreach sessions, the vials of vaccines were routinely placed in polythene bags and transported from peripheral vaccine stores in vaccine carriers. For the present study, a TransTracker C freeze-event indicator was placed in the polythene bag containing the vaccine vials for each outreach session that we investigated. These indicators were checked at the end of the outreach sessions to see if these vials had been exposed to subzero temperatures after they had left the peripheral vaccine stores.
Interviews and store records
In interviews based on a structured questionnaire, the personnel who handled vaccine vials in the cold chain were asked about the maintenance of cold-chain equipment. Records of temperatures that had been routinely kept in each vaccine store were reviewed for the previous 3 months.
At the end of the study, test boxes were collected from the peripheral vaccine stores and transported – at room temperature – to the National Institute of Epidemiology in Chennai. The data from the dataloggers were downloaded and analysed. We calculated the proportions of the 40 boxes that had been exposed to a temperature of < 0 or > 8 °C at each level of the cold chain and while in transit and the percentages of the time that – while stored at each level of the cold chain or in transit – each datalogger had been exposed to a temperature of < 0 or > 8 °C. For each datalogger, we also estimated the longest period of exposure to a temperature of < 0 or > 8 °C, assuming that the datalogger had remained at a similar temperature between two consecutive temperature recordings of < 0 or > 8 °C. The freeze-event indicators in the test boxes were read after the boxes had been transported – unchilled – to Chennai. A shake test was conducted on all DPT vials retrieved from the peripheral vaccine stores.
The study protocol was approved by the institutional ethics committee of the National Institute of Epidemiology. Permission to conduct the study was obtained from the Indian Ministry of Health and Family Welfare and the health authorities in each study state.
The 40 dataloggers recorded temperatures over a total of 138 476 hours. The temperature data recorded while the test boxes were in vaccine stores and in transit accounted for > 99% and < 1% of this total time, respectively.
Proportions of boxes exposed to suboptimal temperatures
During their storage at state and regional vaccine stores, 11% (4/36) and 26% (5/19) of the test boxes were exposed to subzero temperatures while 89% and 58% of the boxes were exposed to temperatures of > 8 °C, respectively (Table 1). The corresponding proportions at peripheral vaccine stores were 63% (25/40) and 88% (35/40), respectively. During their transportation – and depending on the level of the cold chain at which the transport began – 18–36% of the test boxes were exposed to subzero temperatures and 0 to 66% were exposed to temperatures of > 8 °C (Table 1).
Temperatures at vaccine stores
While at state, regional and district vaccine stores, the dataloggers only spent 1.5% (95% confidence interval, CI: 1.4–1.6), 0.2% (95% CI: 0.1–0.2) and 0.6% (95% CI: 0.5–0.7) of their time at subzero temperatures, respectively. Although exposure to such low temperatures was only recorded at one of the state vaccine stores that we investigated – the one for Tamil Nadu (Table 2) – temperatures of > 8 °C were recorded at all of the state stores that we investigated except the one for Tamil Nadu. At regional and district vaccine stores, the dataloggers spent 13.2% and 8.3% of their time at temperatures of > 8 °C, respectively. At the peripheral health facilities, exposure to temperatures of < 0 or > 8 °C was quite common and accounted for 10.5% (95% CI: 10.2–10.8) and 14.7% (95% CI: 14.3–15) of the time that the dataloggers spent in such facilities, respectively. Temperatures of > 8 °C were particularly common in the peripheral vaccine stores in the states of Manipur, Bihar and Madhya Pradesh (Table 3). Overall, temperatures of < 0 and > 8 °C were recorded for 4% and 13% of the combined time –137 519 hours – that the dataloggers recorded temperatures in vaccine stores.
Table 2. Times that vaccines in state and regional stores spent at suboptimal temperatures, India, 2012
Table 3. Times that vaccines in district and peripheral stores and in transit spent at suboptimal temperatures, India, 2012
In some vaccine stores, the members of the research team quickly identified the reasons for the suboptimal temperatures that had been recorded. At the state vaccine store in Tamil Nadu, for example, a thermostat had been turned too low, resulting in subzero temperatures, while gaps between the plates forming the walls of the walk-in cooler used by the state vaccine store in Manipur allowed the temperatures in the cooler to rise above the optimal values.
Temperatures during transit
Temperatures during transit between vaccine stores were recorded for a total of 957 hours. Exposures to temperatures of < 0 and > 8 °C were recorded for 173.5 (18.1%; 95% CI: 15.8–20.8) and 69.5 hours (7.3%; 95% CI: 5.8–9.2) in transit, respectively. Exposure to subzero temperatures was particularly frequent during the shipment of the vaccines in Himachal Pradesh (52.2% of transit time), Manipur (25.9%), Arunachal Pradesh (23.6%) and Bihar (18%), whereas exposure to a temperature of > 8 °C during transit was most common in Bihar (23.3% of transit time) and Andhra Pradesh (16.2%) (Table 3).
Temperatures during outreach sessions
Two of the 40 bags of vaccine vials investigated – both used in Tamil Nadu – were found to have been exposed to subzero temperatures during outreach sessions.
Duration of exposure to suboptimal temperatures
“Continuous spells” – that is, periods that covered at least two consecutive readings by a datalogger – accounted for about 85% of the exposure to subzero temperatures (Table 4) and 94% of the exposure to temperatures of > 8 °C (Table 5) that we recorded.
Table 4. Spells of continuous exposure of vaccines to temperatures of < 0 °C and results of shake tests, India, 2012
Table 5. Spells of continuous exposure of vaccines to temperatures of > 8 °C, India, 2012
Temperature monitoring with freeze-event indicators
Although subzero temperatures were recorded by the dataloggers in 34 of the 40 test boxes, the freeze-event indicators in 31 boxes showed evidence of freezing at any point during the study. The results seen with the two types of freeze-event indicators were identical.
At the end of the study period, 67 of the 80 DPT vials that had been transported to peripheral vaccine stores were sent to the National Institute of Epidemiology for shake tests. Fifty-one (76%) of these 67 vials failed the test, indicating that the vaccines they contained had been frozen during the study period (Table 4).
Routine temperature monitoring and equipment maintenance
In five of the state vaccine stores that we investigated – those in Andhra Pradesh, Bihar, Madhya Pradesh, Tamil Nadu and West Bengal – wireless dataloggers were being used routinely for temperature monitoring. The staff in the other state vaccine stores – and most of those in vaccine stores at lower levels of the cold chain – relied on the temperatures displayed on the control panels of coolers and refrigerators or on dial thermometers. There was no evidence that any of the devices used for routine temperature monitoring in the vaccine stores had ever been calibrated. Although every vaccine store that we investigated could provide records of the temperatures in the stores for the previous 3 months, there was no evidence to show that these records were routinely reviewed to identify any deviations from the recommended temperatures. Almost all of the routine records of temperature indicated values of 2–8 °C.
Maintenance of the cold-chain equipment in the states of Karnataka, Himachal Pradesh and West Bengal was outsourced to the private sector. In the other states that we investigated, such maintenance was the responsibility of cold-chain technicians employed by the states’ health departments. In all of the states, the mean frequency of defrosting of ice-lined refrigerators in primary or community health centres varied from once a week to once a month. The median ages of the walk-in coolers found in state and regional vaccine stores was 12 years (range: 2–27 years) while the median age of the ice-lined refrigerators – which were generally found at lower levels of the cold chain – was 11 years (range: 1–23 years). The irregularity of the power supply was a major issue in all of the facilities that we investigated and especially in the primary or community health centres of Manipur, Bihar, Madhya Pradesh and Tamil Nadu.
In terms of the quantity of vaccines used, numbers of beneficiaries, geographical spread, personnel and cold-chain equipment involved, India has one of the largest immunization programmes in the world.10 Each year, the country spends more than 500 million United States dollars on its national immunization programme. Several new or underused vaccines are likely to be introduced in this programme in the near future. Maintenance of an adequate cold-chain system is essential if vaccines of good quality are to reach each child or woman to be vaccinated. The present study indicates that – as they work their way from state vaccine stores to the children needing them – the vaccines are frequently exposed to temperatures that are lower or higher than the recommended values, with presumably adverse effects on the quality of the vaccines.
During their storage in the vaccine stores in most of the study states, the vials of vaccine in the test boxes were more frequently exposed to temperatures that were higher than the recommended values than to subzero temperatures. The problem of exposure to high temperatures during storage appeared to be particularly acute in the states of Manipur, Madhya Pradesh, Bihar and Himachal Pradesh. In general, exposure to subzero temperatures was found to be common only during storage at peripheral health facilities and in the relatively short periods that the vaccines were in transit between storage facilities. In a systematic review of the results of 14 investigations on cold-chain systems in developing countries, the summary findings were that 21.9% (95% CI: 10.3–33.6) of the refrigerators used for storage and 35.3% (14.8–55.8) of shipments were exposed to freezing temperatures.15 Freezing during transportation is usually the result of the use of ice packs that have not been properly “conditioned” – that is, allowed to warm to 0 °C before use.16,17
In general, vaccine stability and potency are both temperature-dependent. Excessive heat can alter the protein structure and/or chemical stability of a vaccine and result in loss of potency. Lyophilized vaccines such as measles vaccines need to be kept cold and to be used within 4 hours after they have been reconstituted. Otherwise, such vaccines lose stability and are at increased risk of bacterial contamination. Exposure to freezing causes agglomeration of the aluminium salt adjuvants used in DPT and hepatitis vaccines4,18 and – since the size of the granules that form increases on repeated freezing and thawing19 – exposure to freezing probably has a cumulative adverse effect. In laboratory studies, the freezing of a vaccine has been found to cause the antigen to become disassociated from the adjuvant and reduce immunogenicity.3 DPT, diphtheria and tetanus and tetanus toxoid vaccines that have been frozen are more likely to cause local reactions than vaccines that have never been frozen.3 Although there is little direct evidence of vaccine failure in the field as the result of exposure of the vaccine to subzero temperatures, there have been a few reports from Mongolia of lower antibody responses to hepatitis B vaccine as the result of vaccine freezing.20,21
The main aim of the present study was to evaluate the extremes of temperature that occurred in India’s national cold-chain system for vaccines. However, the members of the research team sometimes easily identified the reasons why vaccines were being exposed to suboptimal temperatures – such as an incorrectly set thermostat. The causes of the suboptimal temperatures recorded in all of the vaccine stores need to be identified. There seemed to be a general lack of attention to detail: the devices used for the routine monitoring of the temperatures in vaccine stores had never been calibrated, the temperature measurements that were made routinely were not regularly reviewed, and there was rarely a fixed schedule for the defrosting of refrigerators. Power cuts, shortages of trained personnel and irregular defrosting of refrigerators appeared to be the key issues affecting the maintenance of the cold chain at the district and subdistrict vaccine stores. The evidence for vaccine freezing during transport pointed towards the need for the conditioning of the ice packs used to keep the vaccines cool while in transit.16,17
Our study has certain limitations. Temperatures during the transport of vaccine from the peripheral stores to outreach sessions were only monitored with freeze-event indicators and not with the more informative dataloggers. Although the dataloggers were often exposed to temperatures of > 8 °C, we were not able to determine if such temperatures damaged the vaccine and we made no attempt to test the potency of the vaccine in each of the vials in the test boxes at the end of the study. Unfortunately, all of the DPT vials used in the study had been procured from the state vaccine store in Tamil Nadu – the only state vaccine store where our dataloggers recorded subzero temperatures. Although shake testing of these DPT vials indicated that none had been frozen before the start of our study, it remains possible – and perhaps probable – that they had already been exposed to subzero temperatures at that stage. The high number of DPT vials that failed the shake test at the end of our study may therefore have reflected the cumulative effect of exposures to subzero temperatures in the cold-chain system during our study and similar exposures at the state vaccine store at Tamil Nadu before the study began.
Based on the findings of our study, we propose several recommendations for improving the maintenance of the Indian cold chain. Briefly, there is an urgent need to identify the reasons for the suboptimal temperatures recorded in state, regional and district vaccine stores and to ensure the continuous and accurate monitoring of temperatures in these stores, using digital devices. There is a need to conduct refresher training of the staff of the peripheral stores, with emphasis on the regular defrosting of refrigerators. All staff involved in the transport of vaccines need to be taught how to condition ice packs, and the supervision of the monitoring of cold-chain temperatures needs to be improved. At all levels, there is a need for regular calibration of the devices used to record temperatures. Programme managers at all levels of the cold chain need to review temperature records regularly. The recording of a suboptimal temperature should prompt a rapid corrective response, such as a request for a visit from a cold-chain technician. Although a good cold-chain system may appear expensive, it is likely to be cost-effective, especially whenever costly combination vaccines need to be used in developing countries. Although the development of thermostable vaccines could improve vaccine effectiveness and decrease the need for a cold chain,22,23 such vaccines will probably be expensive, at least when they are initially marketed. Developing countries are therefore likely to need effective cold-chain systems for many more years.
We thank several colleagues for their critical comments on the manuscript, help in the study design or data analysis, and provision of the freeze-event indicators. We also thank the members of the Technical Advisory Group that was created for the study, for their valuable inputs in planning the study. The TransTracker C indicators were kindly provided, free of charge, by Temptime Corporation USA through their distributor Lisaline Lifescience Technologies.
The study was supported financially by the United Nations Children’s Fund in New Delhi.
- World development report: investing in health. New York: Oxford University Press; 1993.
- Study protocol for temperature monitoring in the vaccine cold chain. Geneva: World Health Organization; 2011.
- Temperature sensitivity of vaccines. Geneva: World Health Organization; 2006.
- Galazka A, Milstien J, Zaffran M. Thermostability of vaccines. Geneva: World Health Organization; 1998.
- Boros CA, Hanlon M, Gold MS, Roberton DM. Storage at –3°C for 24 h alters the immunogenicity of pertussis vaccines. Vaccine 2001; 19: 3537-42 http://dx.doi.org/10.1016/S0264-410X(01)00063-9 pmid: 11348721.
- Ho MM, Mawas F, Bolgiano B, Lemercinier X, Crane DT, Huskisson R, et al., et al. Physico-chemical and immunological examination of the thermal stability of tetanus toxoid conjugate vaccines. Vaccine 2002; 20: 3509-22 http://dx.doi.org/10.1016/S0264-410X(02)00342-0 pmid: 12297396.
- Bolgiano B, Mawas F, Yost SE, Crane DT, Lemercinier X, Corbel MJ. Effect of physico-chemical modification on the immunogenicity of Haemophilus influenzae type b oligosaccharide-CRM(197) conjugate vaccines. Vaccine 2001; 19: 3189-200 http://dx.doi.org/10.1016/S0264-410X(01)00024-X pmid: 11312015.
- Diminsky D, Moav N, Gorecki M, Barenholz Y. Physical, chemical and immunological stability of CHO-derived hepatitis B surface antigen (HBsAg) particles. Vaccine 1999; 18: 3-17 http://dx.doi.org/10.1016/S0264-410X(99)00149-8 pmid: 10501230.
- World Health Organization. Expanded Programme on Immunization. The effects of freezing on the appearance, potency, and toxicity of adsorbed and unadsorbed DTP vaccines. Wkly Epidemiol Rec 1980; 55: 385-92.
- Handbook for vaccine and cold chain handlers. New Delhi: Ministry of Health and Family Welfare; 2010. Available from: http://www.unicef.org/india/Cold_chain_book_Final_(Corrected19-04-10).pdf [accessed 20 June 2013].
- Samant Y, Lanjewar H, Parker D, Block L, Tomar GS, Stein B. Evaluation of the cold-chain for oral polio vaccine in a rural district of India. Public Health Rep 2007; 122: 112-21 pmid: 17236617.
- Goel NK, Pathak R, Galhotra A, Dankal C, Swami HM. Status of cold-chain maintenance in Chandigarh. Indian J Public Health 2008; 52: 37-9 pmid: 18700720.
- Galhotra A, Goel NK, Pathak R, Kumar R, Swami HM. Surveillance of cold chain system during intensified pulse polio programme – 2006 in Chandigarh. Indian J Pediatr 2007; 74: 751-3 http://dx.doi.org/10.1007/s12098-007-0132-9 pmid: 17785898.
- Kartoglu U, Ozgüler NK, Wolfson LJ, Kurzatkowski W. Validation of the shake test for detecting freeze damage to adsorbed vaccines. Bull World Health Organ 2010; 88: 624-31 http://dx.doi.org/10.2471/BLT.08.056879 pmid: 20680128.
- Matthias DM, Robertson J, Garrison MM, Newland S, Nelson C. Freezing temperatures in the vaccine cold chain: a systematic literature review. Vaccine 2007; 25: 3980-6 http://dx.doi.org/10.1016/j.vaccine.2007.02.052 pmid: 17382434.
- Training for middle-level managers: cold chain, vaccines and safe-injection equipment management. Geneva: World Health Organization; 2008. Available from: http://www.who.int/immunization_delivery/systems_policy/MLM_module1.pdf [accessed 20 June 2013].
- Nelson C, Froes P, Dyck AM, Chavarría J, Boda E, Coca A, et al., et al. Monitoring temperatures in the vaccine cold chain in Bolivia. Vaccine 2007; 25: 433-7 http://dx.doi.org/10.1016/j.vaccine.2006.08.017 pmid: 17000036.
- Braun LJ, Tyagi A, Perkins S, Carpenter J, Sylvester D, Guy M, et al., et al. Development of a freeze-stable formulation for vaccines containing aluminum salt adjuvants. Vaccine 2009; 27: 72-9 http://dx.doi.org/10.1016/j.vaccine.2008.10.027 pmid: 18973782.
- Menon PS, Sahai G, Joshi VB, Murthy RG, Boparai MS, Thomas AK. Field trial on frozen and thawed tetanus toxoid. Indian J Med Res 1976; 64: 25-32 pmid: 1270098.
- Edstam JS, Dulmaa N, Nymadawa P, Rinchin A, Khulan J, Kimball AM. Comparison of hepatitis B vaccine coverage and effectiveness among urban and rural Mongolian 2-year-olds. Prev Med 2002; 34: 207-14 http://dx.doi.org/10.1006/pmed.2001.0972 pmid: 11817916.
- Edstam JS, Dulmaa N, Tsendjav O, Dambasuren B, Densmaa B. Exposure of hepatitis B vaccine to freezing temperatures during transport to rural health centers in Mongolia. Prev Med 2004; 39: 384-8 http://dx.doi.org/10.1016/j.ypmed.2004.01.029 pmid: 15226050.
- Chen D, Kristensen D. Opportunities and challenges of developing thermostable vaccines. Expert Rev Vaccines 2009; 8: 547-57 http://dx.doi.org/10.1586/erv.09.20 pmid: 19397412.
- Lee BY, Cakouros BE, Assi TM, Connor DL, Welling J, Kone S, et al., et al. The impact of making vaccines thermostable in Niger’s vaccine supply chain. Vaccine 2012; 30: 5637-43 http://dx.doi.org/10.1016/j.vaccine.2012.06.087 pmid: 22789507.